Our Kit is special made to test for diabetes to test for glucose and Ketone in 1 kit
*Blood Glucose Meter
*Blood Ketone Meter
*50 Glucose test strips
*20 Blood ketone test strips
Smart Bluetooth Glucometer
– Fast result: Get blood glucose test result in 10 seconds
– Small blood sample: Only 1µL for blood glucose test
– 460 memories with average reading for 7, 14, 21 and 28 days
– Auto-off power saving
– Low battery indicator
Measurement method : Electro-Chemical Method
Blood sample volume: 1.0 µL
Measure Time : Within 10 seconds
Memory : 460, average reading for 7, 14, 21 and 28 days
Measurement Range : 20 ~ 600 mg/dL (1.1 ~ 33.3 mmol/L)
Size: 82 (L) x 54 (W) x 21.5 (D) mm
Weight : 45g
Automatic Power Off: 1 minute after non use
Battery : 1 x Lithium (R2032)
Accessories: blood sampling device, 10pcs lancets, 10pcs test strips, carrying bag, instruction manual, battery
Blood Ketone Meter SIFKETONE-1.1
SIFKETONE-1.1Blood ketone Monitoring System is used to measure blood ketone
using peripheral blood or vein blood by laypersons (most of them are
affected by diabetes mellitus.) in home or by physician in hospital.
As for diabetics, the energy is shortage when the body can not use glucose normally. The decomposition of fat increases to supply energyand a large amount of ketone is produced correspondingly in the body.It would cause metabolic acidosis or serious life-threatening if much of ketone is accumulated in the body. It is very important for diabetics to test ketone in irregular-time. Not only that, some studies show that blood ketone of diabetics will be significantly increased before the rise of blood glucose levels, which may play the role of early warning for diabetes deterioration.
A small drop of blood, obtained by pricking the skin with a lancet, is placed on a disposable test strip that the meter reads and uses to calculate the
blood Ketone level. The meter then displays the level in units of mg/dl or mmol/l.
The β-HBA, in the presence of hydroxybutyrate dehydrogenase, was
oxidized to acetoacetate with the concomitant reduction of NAD t o NADH.
The NADH was reoxidized to NAD by an electron mediator, and the
current generated was directly proportional to theβ-HBA concentration.
After 30 s, the concentrations (mmol/l) were displayed.
10 to 40
-Protection from diabetic ketoacidosis
-Blood ketone monitoring: 30-second test for ketone body
-Advanced, professional, rapid, convenient and inexpensive;
-Used by the world leading bio-enzyme technology,
-better than another test methods for ketone body;
-Siphon sample injection, automatic switch-on & switch-off;
-Track display for time and date;
-Easy to operate;
-Self-test for those patients with symptom of diabetic ketoacidosis;
-Top choice for severe diabetics;
-Application of ketone test for pregnant women who are succumbed to severe vomiting.
Meter Storage Conditions: -40℃ to 55℃
860～1060hPa atmospheric pressure
System Operating Conditions: 10℃ to 40℃
860～1060hPa atmospheric pressure
Measuring Range (Testing Blood Keonte): 0.3mmol/L~6.0mmol/L
Memory Capacity (Testing Blood Ketone):1 blood ketone results
with time and date
Display: LCD display
Blood volume (Testing Blood Ketone): 2.5uL
Power supply: 1.5V×2DC(Type:AAALR03)
Test strip Storage Conditions: 4℃ to 30℃
Atmospheric pressure 860～1060hPa
Construction: Hand held
Protection Class: Ⅲ
Rated Operation: Continuous operation
The recovery of D-3-hydroxybutyric acid is 80%-120%.
The repeatability using capillary blood with the blood ketone systems is <
0.1mmol/L SD at a level of<1.0mmol/L and <7.5% CV at a level of ≥1.0mmol/L.
Detection limit (lowest value displayed):
0.3mmol/L (3.12mg/dL) for the blood ketone test strips
– 6 years warranty : SIFSOF warrants that your meter will be free from defects in materials
and workmanship for 6 years from the date of purchase. If during this
6-year period, the meter does not work properly because of a defect in
materials or workmanship, will repair the meter or replace it with
ketone meter or equivalent product free of charge. The warranty of the
repaired or the replacement meter will expire on the date of the original
warranty expiration or ninety days after shipment of a replacement
system, whichever period is longer.